Accredited ISO 17025
More PTC® Instruments
PTC® Training Handcuffs
OOO & OOO-S Durometers
PTC® Shore OOO-S Scale Classic Durometer 303
PTC®'s 'Shore Scale' OOOS Durometer Model 303
Shore Scale OOO-S Durometer Model 303
Write a Review
PTC® Shore OOO-S Hardness Scale Classic Durometer
Add Accredited Durometer Certification to ISO 17025
No written calibration certificate required
Add NIST/A2LA Durometer Certification [+$160.00]
Durometer Calibration Intervals? Please Select
Put me on the Waiting List
Add to Cart
Model 303 ASTM type OOO-S Shore scale durometer measures the hardness of viscoelastic polymers. Applications include motorcycle seats, medical pads and wheelchair cushions for pressure ulcer prevention. equestrian pads, gun recoil pads, and cool gel mattresses.
What is the difference between OOO and OOO-S?
There are currently two types of OOO durometers that comply with ASTM D2240 now recognized in the industry. The two distinct OOO units differ in indenter geometry and extension as well as spring force calibration. ASTM D2240 recognizes both variations of the OOO and OOO-S.
For the best accuracy, use of this durometer with one of our accessory Test Stands is recommended. The Model 470 is compatible with this durometer.
Accuracy of ±2 Points (±3.6gf)
2 X 2 inch presser foot
Accessory Test Stands Available
NIST Certification Available
Includes Precision Aluminum Test Block & Carrying Case
Meets or exceeds current ASTM D2240 Specs
Range (Type OOO-S) 0-100 points
Height 4" (10.2cm)
Width 2¼ " (5.7cm)
Depth 1¾ " (4.4cm)
Weight 8 oz (227g)
Shipping Weight 4 lb (1.4kg
For a complete calibration check of mainspring, and visual and mechanical check of indenter, the instrument should be returned to PTC® (see Guarantee & Calibration Service) or refer to ASTM D2240 Specifications. PTC® recommends the unit be returned at least every 12 months for this check. For a quick field check, follow the guideline below. Under no circumstance should a test block be used as a standard to calibrate a durometer.
1. The pointer should read zero when no force is applied to the indenter of the durometer.
2. Hand hold the durometer and insert the indenter into the hole of the calibrated test block. Apply enough force to make firm contact between the top surface of the test block and the base of the durometer. The dial reading should agree with the value stamped on the check block (±1). Several tests should be made and the results averaged.
3. The indenter must protrude 0.098 to 0.100 inches below the base of the durometer.
4. When the indenter is fully displaced, the durometer should read 100 points. Use care as to not damage the tip of the indenter.
PTC® Deadweight Test Stand Durometer OO | OOO 470
Add To Cart
Recently Viewed Items
Turn History Off
The email you entered isn't valid.
You subscribed successfully.