ACCREDITED ISO 17025
Durometer Test Stands
STEEL HARDNESS TESTERS
PTC® TRAINING HANDCUFFS
6" BENCH POLARISCOPE
Melting Point Meter
PTC® OUTLET STORE
OOO & OOO-S Durometers
Durometer Shore OOO PTC® Classic Style #412
ASTM OOO Classic Durometer #412
Durometer Shore OOO PTC® Instruments Classic Style #412
Write a Review
PTC® Shore OOO Scale Classic Durometer
Made in the USA
Rubber Test Blocks ASTM Type OOO
Add Test Blocks for ASTM Type OOO $325.00
A2LA Accredited Calibration Certificate
Add ISO 17025 Durometer Calibration Certificate $195.00
Put me on the Waiting List
Add to Cart
Compatible Test Stands
The PTC® Classic Style Type OOO Durometer comes complete with test block and carrying case. This instrument is designed to be used handheld or with PTC's Deadweight Test Stand model 470.
Shore OOO Scale measures the hardness of very soft materials, suit foams, sponge rubber, open and closed cells, human & animal tissue.
Documentation PDF: 412 Type OOO
Max Hold Pointer Standard
Easy to Read 0 to 100 Point Dial
Accessory Test Stands Available
NIST/A2LA to ISO 17025 Certification Available
Includes Precision Aluminum Test Block
Meets or exceeds current ASTM D2240 Specs
Range (Type OOO) 0-100 points
Accuracy of ±2 Points (±1.8 gf)
Height 4 in. (10.2 cm)
Width 2-1/4 in. (5.7 cm)
Depth 1-3/4 in. (4.4 cm)
Weight 8 oz. (227 g)
Shipping Weight 3lb (1.4kg)
For a complete calibration check of mainspring, and visual and mechanical check of indenter, the instrument should be returned to PTC® (see Guarantee &Calibration Service) or refer to ASTM D2240 Specifications. PTC® recommends the unit be returned at least every 12 months for this check. For a quick field check, follow the guideline below. Under no circumstance should a test block be used as a standard to calibrate a durometer.
1. The pointer should read zero when no force is applied to the indenter of the durometer.
2. Hand hold the durometer and insert the indenter into the hole of the calibrated test block. Apply enough force to make firm contact between the top surface of the test block and the base of the durometer. The dial reading should agree with the value stamped on the check block (±1). Several tests should be made and the results averaged.
3. The indenter must protrude 0.098 to 0.100 inches below the base of the durometer.
4. When the indenter is fully displaced, the durometer should read 100 points. Use care as to not damage the tip of the indenter.
470 Deadweight Test Stand
Durometer Certification to ISO 17025
Add To Cart
Durometer Shore OOO-S PTC® Classic Style #303
Durometer Test Block ASTM Type OOO #404
Durometer Test Stand OO and OOO PTC® #470
Recently Viewed Items
Turn History Off