Accredited ISO 17025
More PTC® Instruments
PTC® Training Handcuffs
Shore D Scale Durometers
PTC® Shore D Scale Durometer #502D
Shore D Durometer Model 502D
PTC® Shore D Scale Durometer Model 502D
Write a Review
PTC® Shore D Scale Analog e2000 Durometer
Need a Durometer Stand? See Compatible Stands Below.
Made in the USA
Add Accredited Durometer Certification to ISO 17025
No written calibration certificate required
Add NIST/A2LA Durometer Certification [+$175.00]
Test Block Kit Type D
Rubber Test Block Kit Type D [+$225.00]
Put me on the Waiting List
Add to Cart
Compatible Test Stands
Easier handling and more repeatable readings are the results of PTC®'s Ergonomic Research. The durometer is easy to grip and the 1.25 diameter base has a large landing area for increased stability and maximum repeatability.
The instrument uses a dial gauge with a full 360º scale with bold, easy to read numbers. The dial scale can easily be read to within a 1/2-point. Accuracy of the instrument is ±1 point. These durometers meet or exceed all aspects of ASTM D2240.
The 502D durometer and the 409D Ergo durometer were chosen as the standard by the Oil & Gas Industry for pipeline coating hardness.
1.25" Diameter Base - The Optimal Size & Shape for Making Measurements
Max Hold Pointer Standard
Easy to Read 0 to 100 Point Dial
Accessory Test Stands Available for all Models
NIST/A2LA to ISO 17025 Certification Available
Includes Precision Aluminum Test Block & Carrying Case
Meets or exceeds current ASTM D2240 Specs
Range (Type A) 0-100 points
Accuracy of ±1 Point
Height 6 in. (15.2 cm)
Width 2-1/4 in. (5.7 cm)
Depth 1-1/2 in. (3.8 cm)
Weight 12 oz. (170 g)
Shipping Weight 3lb (1.4kg)
For a complete calibration check of mainspring, and visual and mechanical check of indenter, the instrument should be returned to PTC® (see Guarantee & Calibration Service) or refer to ASTM D2240 Specifications. PTC® recommends the unit be returned at least every 12 months for this
check. For a quick field check, follow the guideline below. Under no circumstance should a test block be used as a standard to calibrate a durometer.
1. The pointer should read zero when no force is applied to the indenter of the durometer.
2. Hand hold the durometer and insert the indenter into the hole of the calibrated test block. Apply enough force to make firm contact between the top surface of the test block and the base of the durometer. The dial reading should agree with the value stamped on the check block (±1). Several tests should be made and the results averaged.
3. The indenter must protrude 0.098 to 0.100 inches below the base of the durometer.
4. When the indenter is fully displaced, the durometer should read 100 points. Use care as to not damage the tip of the indenter.
Hydraulic Durometer Stand 7000D
Deadweight Durometer Stand 479
Durometer Certification to ISO 17025
Add To Cart
Type D Rubber Test Block Kit #400D
PTC® Stand for C | D | DO e2000 Durometers #479
Add To Cart
PTC® Hydraulic Stand C | D | DO Durometers #7000D
Recently Viewed Items
Turn History Off
The email you entered isn't valid.
You subscribed successfully.