ACCREDITED ISO 17025
STEEL HARDNESS TESTERS
PTC® TRAINING HANDCUFFS
6" BENCH POLARISCOPE
Melting Point Meter
C Scale Durometers
Durometer Shore C PTC® e2000 Style #502C
Durometer Shore C PTC® Instruments e2000 Style #502C
Write a Review
PTC® Shore C Scale Analog e2000 Durometer
Need a Durometer Stand? See Compatible Stands Below.
Made in the USA
Add Accredited Durometer Certificate ISO 17025
No written calibration certificate required
Add NIST/A2LA Durometer Certification
Durometer Calibration Intervals? Please Select
Put me on the Waiting List
Add to Cart
Compatible Test Stands
C Scale Durometers Measure the Hardness of Medium Hard Rubber, Elastomers, and Thermoplastics.
The design of the e2000 is the ultimate in hand held hardness measurements. Easier handling and more repeatable readings are the result of PTC®'s Ergonomic Research. The durometer is easy to grip and the 1.25 diameter base has a large landing area for increased stability and maximum repeatability.
1.25" Diameter Base - The Optimal Size & Shape for Making Measurements
Max Hold Pointer Standard
Easy to Read 0 to 100 Point Dial
Accessory Test Stands Available for all Models
NIST/A2LA to ISO 17025 Certification Available
Includes Precision Aluminum Test Block & Carrying Case
Meets or exceeds current ASTM D2240 Specs
Range (Type C) 0-100 points
Accuracy of ±1 Point
Height 6 in. (15.2 cm)
Width 2-1/4 in. (5.7 cm)
Depth 1-1/2 in. (3.8 cm)
Weight 12 oz. (170 g)
Shipping Weight 3lb (1.4kg)
For a complete calibration check of mainspring, and visual and mechanical check of indenter, the instrument should be returned to PTC® (see Guarantee & Calibration Service) or refer to ASTM D2240 Specifications. PTC® recommends the unit be returned at least every 12 months for this
check. For a quick field check, follow the guideline below. Under no circumstance should a test block be used as a standard to calibrate a durometer.
1. The pointer should read zero when no force is applied to the indenter of the durometer.
2. Hand hold the durometer and insert the indenter into the hole of the calibrated test block. Apply enough force to make firm contact between the top surface of the test block and the base of the durometer. The dial reading should agree with the value stamped on the check block (±1). Several
tests should be made and the results averaged.
3. The indenter must protrude 0.098 to 0.100 inches below the base of the durometer.
4. When the indenter is fully displaced, the durometer should read 100 points. Use care as to not damage the tip of the indenter.
Hydraulic Durometer Stand 7000D
Deadweight Durometer Stand 479
Type D Rubber Test Block Kit #400D
Durometer Test Stand C, D, & DO e2000 PTC® #479
Add To Cart
PTC® Hydraulic Durometer Test Stand for Type C, D & DO #7000D
Recently Viewed Items
Turn History Off