ACCREDITED ISO 17025
MORE PTC® INSTRUMENTS
PTC® TRAINING HANDCUFFS
B Scale Durometers
PTC® Shore B Scale Digital e2000 #511B
Digital e2000 ASTM Type B with ISO 17025 Certification
PTC® Shore B Scale Digital Durometer with ISO 17025 Certification e2000 #511B
Write a Review
PTC® Shore B Scale Digital e2000 Durometer
In Stock. Calibration Certificate Included.
NIST/A2LA to ISO 17025 Calibration Certificate Included
Need a Durometer Stand? See Compatible Stands Below.
Add Digital Durometer SmartCable™
No Smart Cable Required
6 Foot Cable [+$345.00]
20 Foot Cable [+$395.00]
Hand Switch or Foot Switch for Smart Cable™
No Switch Required
Hand Switch [+$75.00]
Foot Switch [+$75.00]
Put me on the Waiting List
select name from list
other, add new name below
or add name:
*We'll ask for shipping info at checkout
Add to Cart
Compatible Test Stands
The Digital Durometer features a large direct read LCD display, electronic max hand, USB data ports for computer compatibility, a locking feature to prevent accidental erasure of tolerances, automatic shutoff, and low battery warning.
All PTC® digital durometers come complete with Test Block, Carrying Case, an A2LA/NIST Traceable Calibration Certificate to ISO 17025 from
PTC®'s SmartCable can be added to any digital durometer. The output of the SmartCable USB Keyboard is a standard 2.0 keyboard signal. Readings can be sent to any Windows application's cursor point or highlighted cell. Works great with
. No special software required.
NIST/A2LA to ISO 17025 Certification Included
1.25" Diameter Base - The Optimal Size & Shape for Making Measurements
Max Hold Pointer Standard
LCD Display with 0.1 Point Resolution
Accessory Test Stands Available for all Models
Includes Precision Aluminum Test Block, Batteries, 115V AC Adaptor & Carrying Case
Meets or exceeds current ASTM D2240 Specs
Range (Type B) 0-100 points
Accuracy of ±1 Point
Height 6 in. (15.2 cm)
Width 2-1/4 in. (5.7 cm)
Depth 1-1/2 in. (3.8 cm)
Weight 13 oz. (170 g)
Shipping Weight 4lb (1.4kg)
For a complete calibration check of mainspring, and visual and mechanical check of indenter, the instrument should be returned to PTC® (see Guarantee & Calibration Service) or refer to ASTM D2240 Specifications. PTC® recommends the unit be returned at least every 12 months for this check. For a quick field check, follow the guideline below. Under no circumstance should a test block be used as a standard to calibrate a durometer.
1. The pointer should read zero when no force is applied to the indenter of the durometer.
2. Hand hold the durometer and insert the indenter into the hole of the calibrated test block. Apply enough force to make firm contact between the top surface of the test block and the presser foot. The dial reading should agree with the value stamped on the check block (±1). Several tests should be made and the results averaged.
3. The indenter must protrude 0.098 to 0.100 inches below the presser foot.
4. When the indenter is fully displaced, the durometer should read 100 points. Use care as to not damage the tip of the indenter.
Hydraulic Test Stand 7000A
Deadweight Test Stand 478
Type A Test Block Kit #400A
Digital Durometer USB SmartCable® #511.600
PTC® Stand for A | B | E | O Durometers #478
PTC® Hydraulic Stand A | B | O Durometers #7000A
Aluminum Block ASTM Type B #401B
PTC® Shore B Scale e2000 #501B
Browse Similar Items
Digital e2000 Durometer
Recently Viewed Items
Turn History Off
The email you entered isn't valid.
You subscribed successfully.