ACCREDITED ISO 17025
MORE PTC® INSTRUMENTS
PTC® TRAINING HANDCUFFS
B Scale Durometers
PTC® Shore B Scale Ergo Durometer #408B
Ergo Durometer ASTM Type B
Durometer Shore B Scale with Accuracy of ± 1 Point Model 408B
Write a Review
PTC® Shore B Scale Ergo Durometer
Add a 1kg Load Weight or Durometer Stand
Add 1 kg Load Weight Model 441
Provides a constant load for hand held measurements [+$175.00]
Add Accredited Durometer Certification to ISO 17025
No written calibration certificate required
Add NIST/A2LA Durometer Certification [+$175.00]
Durometer Calibration Intervals? Please Select
Put me on the Waiting List
Add to Cart
Compatible Test Stands
B Scale Durometers Test the Hardness of Moderately Hard Rubber, Elastomers, Paper Products, Fibrous Materials.
Base Tapers to 1" x 1"
Max Hold Pointer Standard
Easy to Read 0 to 100 Point Dial
Accessory Test Stands Available
NIST/A2LA to ISO 17025 Certification Available
Includes Precision Aluminum Test Block
Meets or exceeds current ASTM D2240 Specs
Range (Type B) 0-100 points
Accuracy of ±1 Point
Height 4 in. (10.2 cm)
Width 2-1/4 in. (5.7 cm)
Depth 1-3/4 in. (4.4 cm)
Weight 6 oz. (170 g)
Shipping Weight 3lb (1.4kg)
For a complete calibration check of mainspring, and visual and mechanical check of indenter, the instrument should be returned to PTC® (see Guarantee & Calibration Service) or refer to ASTM D2240 Specifications. PTC® recommends the unit be returned at least every 12 months for this check. For a quick field check, follow the guideline below. Under no circumstance should a test block be used as a standard to calibrate a durometer.
1. The pointer should read zero when no force is applied to the indenter of the durometer.
2. Hand hold the durometer and insert the indenter into the hole of the calibrated test block. Apply enough force to make firm contact between the top surface of the test block and the base of the durometer. The dial reading should agree with the value stamped on the check block (±1). Several tests should be made and the results averaged.
3. The indenter must protrude 0.098 to 0.100 inches below the base of the durometer.
4. When the indenter is fully displaced, the durometer should read 100 points. Use care as to not damage the tip of the indenter.
320 Spring Load Cell Stand
471 1 kg Weight Stand
7000A Hydraulic Stand
PTC® Durometer's Constant Load Weight 1kg #441
Add To Cart
Durometer Certification to ISO 17025
Add To Cart
Type A Test Block Kit #400A
PTC® Durometer Stand for A | B | O Durometers #471
Durometer Stand Spring Load Cell 320
PTC® Hydraulic Stand A | B | O Durometers #7000A
Recently Viewed Items
Turn History Off
The email you entered isn't valid.
You subscribed successfully.