ACCREDITED ISO 17025
MORE PTC® INSTRUMENTS
PTC® TRAINING HANDCUFFS
Shore A Scale Durometers
PTC® Shore A Digital e2000 #511A
Digital Durometer Shore A Scale Model 511A
PTC® Shore A Digital Durometer e2000 #511A
Write a Review
PTC® Shore A Scale Digital e2000 Durometer
In Stock. Calibration Certificate Included.
NIST/A2LA to ISO 17025 Calibration Certificate Included
Need a Durometer Stand? See Compatible Stands Below.
Add Digital Durometer SmartCable™
No Smart Cable Required
6 Foot Cable [+$345.00]
20 Foot Cable [+$395.00]
Hand Switch or Foot Switch for Smart Cable™
No Switch Required
Hand Switch [+$75.00]
Foot Switch [+$75.00]
Put me on the Waiting List
Add to Cart
Compatible Test Stands
The Digital Durometer features a large direct read LCD display, electronic max hand, USB data ports for computer compatibility, a locking feature to prevent accidental erasure of tolerances, automatic shutoff, and low battery warning.
All PTC® digital durometers come complete with Test Block, Carrying Case and A2LA/NIST Traceable Calibration Certificate to ISO 17025 from PTC Metrology™
PTC®'s SmartCable can be added to any digital durometer. The output of the SmartCable USB Keyboard is a standard 2.0 keyboard signal. Readings can be sent to any Windows application's cursor point or highlighted cell. Works great with Mircosoft® Excel. No special software required.
Specifications 511A Digital Durometer
NIST/A2LA to ISO 17025 Certification Included
1.25" Diameter Base - The Optimal Size & Shape for Making Measurements
Max Hold Pointer Standard
LCD Display with 0.1 Point Resolution
Accessory Test Stands Available for all Models
Includes Precision Aluminum Test Block, Batteries, 115V AC Adaptor & Carrying Case
Meets or exceeds current ASTM D2240 Specs
Range (Type A) 0-100 points
Accuracy of ±1 Point
Height 6 in. (15.2 cm)
Width 2-1/4 in. (5.7 cm)
Depth 1-1/2 in. (3.8 cm)
Weight 13 oz. (170 g)
Shipping Weight 4lb (1.4kg)
For a complete calibration check of mainspring, and visual and mechanical check of indenter, the instrument should be returned to PTC® (see Guarantee & Calibration Service) or refer to ASTM D2240 Specifications. PTC® recommends the unit be returned at least every 12 months for this check. For a quick field check, follow the guideline below. Under no circumstance should a test block be used as a standard to calibrate a durometer.
1. The pointer should read zero when no force is applied to the indenter of the durometer.
2. Hand hold the durometer and insert the indenter into the hole of the calibrated test block. Apply enough force to make firm contact between the top surface of the test block and the presser foot. The dial reading should agree with the value stamped on the check block (±1). Several tests should be made and the results averaged.
3. The indenter must protrude 0.098 to 0.100 inches below the presser foot.
4. When the indenter is fully displaced, the durometer should read 100 points. Use care as to not damage the tip of the indenter.
Hydraulic Test Stand 7000A
Deadweight Durometer Test Stand 478
PTC® Stand for A | B | E | O Durometers #478
Digital Durometer USB SmartCable® #511.600
Type A Test Block Kit #400A
PTC® Hydraulic Stand A | B | O Durometers #7000A
Browse Similar Items
Digital e2000 Durometer
Recently Viewed Items
Turn History Off
The email you entered isn't valid.
You subscribed successfully.