Durometer Calibration for Oil, Gas & Coatings Industry
ACCREDITED ISO 17025
MORE PTC® INSTRUMENTS
PTC® TRAINING HANDCUFFS
Foam & Sponge Rubber Durometer
PTC® Foam & Sponge Rubber Durometer #302SL
PTC® Foam & Sponge Rubber Durometer 302SL
PTC® Foam & Sponge Rubber Durometer Model 302SL
Write a Review
PTC® Foam & Sponge Rubber Durometer
Add Accredited Durometer Certification to ISO 17025
No written calibration certificate required
Add NIST/A2LA Durometer Certification [+$175.00]
Durometer Calibration Intervals? Please Select
Put me on the Waiting List
Add to Cart
The Model 302SL Durometer Measures the Hardness of soft materials such as foam rubber, sponge rubber, cellular material and gel-like materials.
This instrument can perform comparative hardness tests on foam or sponge rubber without the cost of using an indention load deflection (ILD) test machine. The PTC® Model 302SL offers a cost-effective means of checking these materials with speed, portability, and accuracy. The Model 302SL is based on the force of a variable load over an area of one square centimeter.
This instrument is simple to use and can be used in conjunction with the Model 470 Test Stand for faster and more precise testing. Several readings should be made and the results averaged. The sample should have an area of at least 1-1/4" square and be at least 3/4" thick. Thinner materials must be stacked (do not glue) to obtain this thickness.
We also offer material testing to determine the correct durometer type and style for your application at no charge.
Max Hold Pointer Standard
Easy to Read 0 to 100 Point Dial
Accessory Test Stands Available
NIST Certification Available
Test Block & Carrying Case (included) Model 405
Accurately Measure the Hardness of Foam, Sponge Rubber
Accuracy of ±2.8 gf
Range (Type SL) 0-100 points
Height 4" (10.2cm)
Width 2¼ " (5.7cm)
Depth 1¾ " (4.4cm)
Weight 9.5 oz (227g)
Shipping Weight 3 lb (1.4kg)
PTC® Deadweight Test Stand Durometer OO | OOO #470
Aluminum Test Block Type SL #405
Recently Viewed Items
Turn History Off
The email you entered isn't valid.
You subscribed successfully.